OMRON IA Global
Digital Panel Meters
Find out information related to Digital Panel Meters.
RoHS Compliant Status
Discontinuation and Replacement
View frequently asked questions in your country.
What is the measuring accuracy when measuring circumferential speed with the K3NR?
Can one sensor with a 4- to 20-mA output be connected to two K3NX or K3TS Process Meters?
How can a DC current of 1,000 A be measured with a K3NX Process Meter?
The BCD output is not being output from a Meter with a BCD output. What should I do?
Which parameter should be set in order to display the K3NRfs rotation speed?
There is no response to commands sent from the host computer to the Meter via serial communications. What should I do?
Is it possible to connect two of the K3NR, K3NP, or K3NC Meters to a single open-collector output sensor?
The sensor is operating normally, but the K3NRfs display is not changing. Why is this happening?
What is the K3NX Meterfs display accuracy?
What is Averaging?
How can I change the Digital Panel Meterfs scaling value from the host computer?
Are there any functions that can prevent output chattering?
What is Simple Average?
What is Moving Average?
What steps can be taken to reduce noise?
What kind of functions does the ThermoTools provide?
Can the ThermoTools be connected to a USB port?
Which products can the ThermoTools be used with?